Stereo Compatible. Coupler Magnification: 0.35X. For SZ0502, SZ0503 Series Trinocular Zoom Body.
Coupler and Camera AdapterClose Λ
|Coupler or camera adapter, is also called photo eyepiece, CCTV adapter, CCD adapter, camera adapter, digital camera adapter, and camera adapter etc.|
Coupler is an optical imaging lens that uses mechanical adapter device to connect the camera to the microscope, and project the image onto the camera's target through the adapter.
Coupler/C-mount adapters have different magnifications to capture images of different magnifications and field of view. In optical imaging, optical parameters such as chromatic aberration, distortion, and field curvature are also to be considered to ensure image quality. At the same time, for some other couple/C-mount adapters, factors such as the position and angle of the camera are also considered to suit the needs of the observational environment.
Coupler/C-mount AdapterClose Λ
|Coupler/C-mount adapter is an adapter commonly used for connection between the C-adapter camera (industrial camera) and a microscope.|
Adjustable CouplerClose Λ
|On the coupler/C-mount-adapter, there is an adjustable device to adjust the focal length.|
Coupler for Microscope TypeClose Λ
|Different coupler/C-mount-adapters are suitable for different microscopes. For some, some adapter accessories need to be replaced. See the applicable range of each coupler/C-mount-adapter for details.|
Coupler MagnificationClose Λ
|Coupler magnification refers to the line field magnification of the coupler/C-mount-adapter. With different magnifications of the adapter lens, images of different magnifications and fields of view can be obtained. The size of the image field of view is related to the sensor size and the coupler/C-mount-adapter magnification.|
Camera image field of view (mm) = sensor diagonal / coupler/C-mount-adapter magnification.
For example: 1/2 inch sensor size, 0.5X coupler/C-mount-adapter coupler, field of view FOV (mm) = 8mm / 0.5 = 16mm.
The field of view number of the microscope 10X eyepiece is usually designed to be 18, 20, 22, 23mm, less than 1 inch (25.4mm). Since most commonly used camera sensor sizes are 1/3 and 1/2 inches, this makes the image field of view on the display always smaller than the field of view of the eyepiece for observation, and the visual perception becomes inconsistent when simultaneously viewed on both the eyepiece and the display. If it is changed to a 0.5X coupler/C-mount-adapter, the microscope image magnification is reduced by 1/2 and the field of view is doubled, then the image captured by the camera will be close to the range observed in the eyepiece.
Some adapters are designed without a lens, and their optical magnification is considered 1X.
For Camera Sensor SizeClose Λ
|For the size of the lens field of view of the coupler/C-mount-adapter, in the design process, the size of the camera sensor imaging target should be considered. When the field of view of the lens is smaller than the target plane of the camera, “black border” and “dark corner” will appear.|
The general microscope coupler/C-mount adapters are generally designed for the 1/2" camera targets. When a camera of 2/3 or larger target is used, the “dark corner” phenomenon will appear in the field of view. Especially, at present, DSLR cameras generally use large target plane design (1 inch full field of view), when used for microscopic photographing, the general DSLR camera coupler/C-mount adapter will have “black border”.
Generally, the “dark corner” that appears on the field of view is often that the center of the microscope and the camera are not aligned. Adjust the position of the screw on the camera adapter, or turn the camera adapter to adjust or change the effect.
C/CS-Mount CouplerClose Λ
|At present, the coupler/C-mount adapter generally adopts the C/CS-Mount adapter to match with the industrial camera. For details, please refer to "Camera Lens Mount".|
|After unpacking, carefully inspect the various random accessories and parts in the package to avoid omissions. In order to save space and ensure safety of components, some components will be placed outside the inner packaging box, so be careful of their inspection.|
For special packaging, it is generally after opening the box, all packaging boxes, protective foam, plastic bags should be kept for a period of time. If there is a problem during the return period, you can return or exchange the original. After the return period (usually 10-30 days, according to the manufacturer’s Instruction of Terms of Service), these packaging boxes may be disposed of if there is no problem.
|Packaging Type||Carton Packaging|
|Packaging Material||Corrugated Carton|
|Packaging Dimensions(1)||4.5x4.5x6.5cm (1.771x1.771x2.559″)|
|Inner Packing Material||Plastic Bag|
|Ancillary Packaging Materials||Sponge|
|Gross Weight||0.08kg (0.18lbs)|
|Minimum Packaging Quantity||1pc|
|Transportation Carton||Carton Packaging|
|Transportation Carton Material||Corrugated Carton|
|Transportation Carton Dimensions(1)||15.2x15.2x15.2cm (6x6x6″)|
|Total Gross Weight of Transportation(kilogram)||0.23|
|Total Gross Weight of Transportation(pound)||0.51|