Finite. Total Magnification: 8-50X. 10X Adjustable Eyepiece. 1X Built-in Objective. Standard Coupler: 0.4X. Zoom Ratio: 1:6.3. Trinocular. Magnification Detent : 1X per pre-set stop. Eye Tube Angle: 45°. Eyepiece Field of View: Dia. 22mm. Stand Type: Boom Stand.
Finite. Total Magnification: 8-50X. 10X Adjustable Eyepiece. 1X Built-in Objective. Standard Coupler: 0.4X. Zoom Ratio: 1:6.3. Trinocular. Magnification Detent : 1X per pre-set stop. Eye Tube Angle: 45°. Eyepiece Field of View: Dia. 22mm. Stand Type: Boom Stand.
SZ02030441 Trinocular Zoom Stereo Microscope
Optical System Specifications
|System Optical Magnification||8-50X|
|Expandable System Optical Magnification (Optional Parts Required)||4-200X|
|Trinocular Optical Magnification||0.32-2X|
|Standard Eyepiece||10X Adjustable Eyepiece|
|Standard Objective||1X Built-in Objective|
|System Field of View||Dia. 4.4-27.5mm|
|Expandable System Field of View||Dia. 1-50mm|
|System Working Distance||115mm|
|Expandable System Working Distance||45-220mm|
Trinocular Zoom Body
|Body Optical System||Finite|
|Zoom Operating Mode||With Two Horizontal Knobs|
|Body Mounting Size for Stand||Dia. 76mm|
|Magnification Detent||1X per pre-set stop|
|Body Mount Type for Coupler||Thread Screw|
|Body Mount Size for Coupler||Dia. 26mm|
|Nosepiece Adapter Size for Ring Light||Dia. 60mm|
|Eye Tube Adjustment Mode||Compensating|
|Eye Tube Angle||45°|
|Erect/Inverted Image||Erect image|
|Eye Tube Rotatable||360° Degree Rotatable|
|Eye Tube Inner Diameter||Dia. 30mm|
|Eye Tube Fixing Mode||Elastic Tube locking|
|Image Port Switch Mode||20/80 True-Trinocular|
|Eyepiece Type||Adjustable Eyepiece|
|Eyepiece Optical Magnification||10X|
|Plan Eyepiece||Plan Eyepiece|
|Eyepiece Size for Eye Tube||Dia. 30mm|
|Eyepiece Field of View||Dia. 22mm|
|Eyepoint Type||High Eyepoint Eyepiece|
|Eyepiece Size for Reticle||Dia. 27mm|
|Eyepiece Diopter Correction||±5°|
|Objective Working Distance||115mm|
|Objective Screw Thread||M55x0.75mm|
|Surface Treatment||Spray Paint|
|Net Weight||1.72kg (3.79lbs)|
|Stand Type||Boom Stand|
|Vertical Post Height||384mm|
|Maximum Vertical Post Extended Length||254mm|
|Vertical Post Diameter||Dia. 37.2mm|
|Cross Adapter Type||Cross Hole Adapter|
|Horizontal Arm Type||Horizontal Post|
|Horizontal Arm Length||544mm|
|Horizontal Diameter||Dia. 37.2mm|
|Mounting Hole on the Top of Horizontal Arm||5/8 in. End Adapter|
|Horizontal Rotation Angle||360° Degree Rotatable|
|Horizontal Arm Travel Distance on Z-Axis||266mm|
|Horizontal Arm Stretch Range||480mm|
|Horizontal Arm Maximum Load||9.50kg (20.94lbs)|
|Horizontal Arm Travel Mode on Horizontal Direction||Manual|
|Horizontal Arm Travel Mode on Z Direction||Manual|
|Base Type||Heavy Duty Base|
|Surface Treatment||Electroplating Black|
|Net Weight||18.14kg (39.99lbs)|
|Dimensions||285x260x600mm (11.220x10.236x23.622 in. )|
|Holder Adapter Type||Dia. 76mm Scope Holder|
|Coarse Focus Distance per Rotation||20mm|
|E-Arm Rotation Range on Horizontal Direction||360°|
|E-Arm Rotation Range on Z Direction||180°|
|E-Arm Mounting Adapter||5/8 in. End Adapter|
|Center Distance from E-Arm Adapter to Scope Holder||130mm|
|E-Arm Horizontal Adjustment Screw||Horizontal Adjustable|
|Surface Treatment||Spray Paint|
|Net Weight||0.84kg (1.85lbs)|
|Coupler Mount Type for Trinocular||Thread Screw|
|Coupler Mount Size for Trinocular||Dia. 26mm|
|Coupler for Microscope Type||Stereo Compatible|
|For Camera Sensor Size||Under 2/3 in.|
|Surface Treatment||Electroplating Black|
|Net Weight||0.06kg (0.13lbs)|
|Applied Field||For SZ17011141, SZ17011142 Trinocular Zoom Body|
|Surface Treatment||Spray Paint|
|Net Weight||20.7kg (45.10lbs)|
|This Kit Includes||SZ17011141, SZ17016121, ST02051101, SA02021102|
Stereo MicroscopeClose Λ
|Stereo microscopes, also known as the anatomical microscope, or dissecting microscope. Many people used to call the stereo microscope Stereo, and the continuous zoom microscope is simply called Zoom.|
Stereo microscope is a kind of binocular microscope that observes an object with both eyes from different angles and thereby causing a stereoscopic effect.
The stereo microscope adopts two independent optical paths, and the left and right beams in the binocular tube have a certain angle, generally 12°~15°, and the objects are observed from different angles of the two optical paths, causing the eyes to have a three-dimensional effect, and therefore stereo microscope is a true 3D microscope.
Compared with other compound microscopes, stereo microscopes belong to the low power optical microscope. The field of view of stereo microscope has a large diameter, its magnification is generally below 200X for optical amplification. When the multiple is greater than 40X, basically the stereoscopic effect of the image will be relatively poor.
Therefore, the advantage of the stereo microscope is not that its magnification is large, but that its working distance is long and the depth of field is large, which is particularly suitable for observing the objects having a high degree of three-dimensional features.
For compound microscope with a single optical path, what we see is only a flat image. Although most compound microscopes have two eyepieces, what we actually see is one and the same image, and this is just to facilitate the observation habits of our two eyes. The stereo microscope has two optical paths (two objective lenses or one common objective lens), and only the three-dimensional sense produced under observation of the two optical paths can make people judge the three-dimensional spatial position of the observed object, which can generate a sense of distance under the microscope. Therefore, only stereo microscope can be used for operation under the microscope which is very suitable for surgery, dissection, industrial welding, assembly, precision instrument repair and so on.
The stereo microscope can be equipped with a wide range of accessories. It can be combined with various digital cameras and photographic interfaces, microscope cameras, eyepiece camera and image analysis software to form a digital imaging system. It can be connected to a computer for analysis and processing, and its lighting system has also different options for illumination, such as reflected light, transmitted light, etc.
Stereoscopic microscopes are widely used in various fields, such as biology, medicine, agriculture, forestry, marine life, and various departments, especially in the industry, which are widely used in macroscopic surface observation, analysis, and microscopic operations.
Stereoscopic microscopes were invented by American instrument engineer Horatio S. Greenough in the 1890s, manufactured by Carl Zeiss Company of Germany, and are widely used in scientific research, archaeological exploration, industrial quality control and biopharmaceuticals etc.
Stereo Microscope Quick Operation Steps
In the working position, place the microscope on the workbench after installation.
Connect the power source, and turn on the light source.
Place an observation type (also known as specimen or sample) such as a coin etc. under the microscope or on the base.
Adjust the focus knob of the stand, measure the height visually or based on the working distance parameters of the objective lens used.
Adjust the zoom knob of the microscope to the lowest multiple. Find the approximate image by adjusting the focus knob lift set, and place a certain feature point of the object that needs to be observed of its image to the approximate center position.
Align the feature points of the specimen, gradually adjust to a large multiple, adjust the lift set of the microscope appropriately, and finally find the focal plane of the maximum multiple. During the adjustment process, use the more obvious feature points on the observed object (coin) to compare the sharpness of the image.
Turn the zoom knob again to the lowest multiple. It is possible that the image may be some how out of focus. At this time, please don't adjust the lift set to focus again. Simply adjust the diopters on the two eyepieces to accommodate eye observations (diopter varies from person to person).
Adjust the viewing distance of the eyepiece to achieve a comfortable position.
At this point, the microscope is already parfocal, i.e., the microscope is changed from high power to low power, and the entire image is in the focal plane. For the same sample to be observed, we do not need to adjust again the other parts of the microscope. We only need to turn the zoom knob to zoom in on the specimen for observation.
Adjust the light source, including adjusting the brightness and angle of incidence of the light source to get the best image, or see more different details.
Adjust other necessary and needed equipment, such as the photographic eyepieces, cameras, etc., to image on the display or to find the sharpest image.
When binocular observation of the left and right images or sharpness is not the same, first adjust the vision adjustment knob on the eyepiece, adjust the parallax of the two eyes, so that the image of the two eyes are consistent. It is normal to feel fatigue when using a microscope for a long time. Take a break and then observe and work again to adapt the human eye to the use of the microscope. If the microscope is used for too long, or if there is a problem inside the microscope due to large temperature difference, vibration, etc., please contact the dealer or contact our service staff on the BoliOptics website.
|Microscopes and components have two types of optical path design structures.|
One type is finite optical structural design, in which light passing through the objective lens is directed at the intermediate image plane (located in the front focal plane of the eyepiece) and converges at that point. The finite structure is an integrated design, with a compact structure, and it is a kind of economical microscope.
Another type is infinite optical structural design, in which the light between the tube lens after passing the objective lens becomes "parallel light". Within this distance, various kinds of optical components necessary such as beam splitters or optical filters call be added, and at the same time, this kind of design has better imaging results. As the design is modular, it is also called modular microscope. The modular structure facilitates the addition of different imaging and lighting accessories in the middle of the system as required.
The main components of infinite and finite, especially objective lens, are usually not interchangeable for use, and even if they can be imaged, the image quality will also have some defects.
The separative two-objective lens structure of the dual-light path of stereo microscope (SZ/FS microscope) is also known as Greenough.
Parallel optical microscope uses a parallel structure (PZ microscope), which is different from the separative two-object lens structure, and because its objective lens is one and the same, it is therefore also known as the CMO common main objective.
System Optical MagnificationClose Λ
|The magnification of the objective lens refers to the lateral magnification, it is the ratio of the image to the real size after the original image is magnified by the instrument. This multiple refers to the length or width of the magnified object.|
System optical magnification is the product of the eyepiece and the objective lens (objective lens zoom set) of the optical imaging part within the system.
Optical magnification = eyepiece multiple X objective lens/objective lens set
The maximum optical magnification of the microscope depends on the wavelength of the light to which the object is illuminated. The size of the object that can be observed must be greater than the wavelength of the light. Otherwise, the light cannot be reflected or transmitted, or recognized by the human eye. The shortest wavelength of ultraviolet light is 0.2 microns, so the resolution of the optical microscope in the visible range does not exceed 0.2 microns, or 200 nanometers. This size is converted to the magnification of the microscope, and it is the optical magnification of 2000X. Usually, the compound microscope can achieve 100X objective lens, the eyepiece is 20X, and the magnification can reach 2000X. If it is bigger, it will be called "invalid magnification", that is, the image is large, but the resolution is no longer increased, and no more details and information can be seen.
Trinocular Optical MagnificationClose Λ
|When the instrument is conducting electronic image magnification and observation through a camera or the like, the optically magnified portion may not be the optical path that passes through the "eyepiece-objective lens" of the instrument, at this time, the calculation method of the magnification is related to the third-party photo eyepiece passed. |
The trinocular optical magnification is equal to the multiplier product of objective lens (objective lens set) and the photo eyepiece
Trinocular optical magnification = objective lens X photo eyepiece
Total MagnificationClose Λ
|Total magnification is the magnification of the observed object finally obtained by the instrument. This magnification is often the product of the optical magnification and the electronic magnification.|
When it is only optically magnified, the total magnification will be the optical magnification.
Total magnification = optical magnification X electronic magnification
Total magnification = (objective X photo eyepiece) X (display size / camera sensor target )
System Field of ViewClose Λ
|Field of View, is also called FOV. |
The field of view, or FOV, refers to the size of the object plane (i.e., the plane of the point of the observed object perpendicular to the optical axis), or of its conjugate plane (i.e., object to primary image distance), represented by a line value.
System field of view is the size of the actual diameter of the image of the terminal display device of the instrument, such as the size of the image in the eyepiece or in the display.
Field of view number refers to the diameter of the field diaphragm of the objective lens, or the diameter of the image plane formed by the field diaphragm.
Field of view number of objective lens = field of view number of eyepiece / (objective magnification / mechanical tube length)
Large field of view makes it easy to observe the full view and more range of the observed object, but the field of view (FOV) is inversely proportional to the magnification and inversely proportional to the resolution, that is, the larger the field of view, the smaller the magnification, and also the lower the resolution of the object to be observed.
There are usually two ways to increase the field of view, one is to replace with an objective lens of a smaller multiple, or to replace with an eyepiece of a smaller multiple.
System Working DistanceClose Λ
|Working distance, also referred to as WD, is usually the vertical distance from the foremost surface end of the objective lens of the microscope to the surface of the observed object.|
When the working distance or WD is large, the space between the objective lens and the object to be observed is also large, which can facilitate operation and the use of corresponding lighting conditions.
In general, system working distance is the working distance of the objective lens. When some other equipment, such as a light source etc., is used below the objective lens, the working distance (i.e., space) will become smaller.
Working distance or WD is related to the design of the working distance of the objective lens. Generally speaking, the bigger the magnification of the objective lens, the smaller the working distance. Conversely, the smaller the magnification of the objective lens, the greater the working distance.
When it is necessary to change the working distance requirement, it can be realized by changing the magnification of the objective lens.
Trinocular Zoom BodyClose Λ
|Trinocular zoom body is the main body of a stereo microscope that has continuous zooming functions. In addition to the two eyepieces for observation, there is a third optical path (image port), which is usually a set of optical paths borrowed from the microscope for connecting to the camera to facilitate the observation with the display or connecting to a computer. Usually, the third ocular of the body can be configured with different photo eyepieces, or other interfaces to connect to different webcam, cameras and so on.|
This body usually needs to be placed on a microscope stand for use. Generally, a variety of eyepieces and objective lenses with different magnifications can be selected, and high-end stereo microscope usually has a wide range of accessories for selection.
Zoom RangeClose Λ
|Zoom in zoom microscope means to obtain different magnifications by changing the focal length of the objective lens within a certain range through adjustment of some lens or lens set while not changing the position of the object plane (that is, the plane of the point of the observed object perpendicular to the optical axis) and the image plane (that is, the plane of the image imaging focus and perpendicular to the optical axis) of the microscope. |
Zoom range refers to the range in which the magnification is from low to high. In the zoom range of the microscope, there is no need to adjust the microscope knob for focusing, and ensure that the image is always clear during the entire zoom process.
The larger the zoom range, the stronger the adaptability of the range for microscope observation, but the image effects at both ends of the low and high magnification should be taken into consideration, the larger the zoom range, the more difficult to design and manufacture, and the higher the cost will be.
Zoom RatioClose Λ
|Zoom ratio is the ratio of the maximum magnification / the minimum magnification. Expressed as 1: (ratio of maximum magnification / minimum magnification). If the maximum magnification is 4.5X, the minimum magnification is 0.7X, then the zoom ratio = 4.5 / 0.7 = 6.4, the zoom ratio will be 1:6.4.|
Zoom ratio is obtained by the intermediate magnification group of the microscope. When the magnification is increased or decreased by using other objective lenses, the zoom ratio does not change accordingly.
With Two Horizontal KnobsClose Λ
|When microscope body changes the magnification, it is realized by adjusting the horizontally placed zoom knob. Because the knob is relatively small, it is therefore easier to zoom and the image is stable.|
For most of the dual stereo microscopes, magnification is realized by adjusting the zoom drum or nosepiece below. When the nosepiece is relatively big, frequent operation is more laborious. Magnifying while observing, the microscope may shake, thereby causing eye discomfort for observation.
Using zoom drum or nosepiece type microscope, if there is a ring light under the microscope, the ring light carries the wire, and when magnification conversion is often required, the ring light and the wire will swing along with the magnification, which makes the operation inconvenient. This situation will not occur to zoom with two horizontal knobs.
Magnification Detent Close Λ
|In the body of zoom microscope, zooming is continuous. When rotating to a certain position, generally an integral multiple, a positioning structure or detent is added, which has a distinct hand feel during the zooming process, and stops at this position.|
When measuring, or testing by factory for unified standard magnification, a magnification detent device can avoid the error caused by the inaccurate multiple positioning of the optical magnification.
|For compensating eyetube, when changing the interpupillary distance, it requires two hands to operate at the same time, with one hand fixing one eyepiece tube, and the other pushing or pulling the other, or both the left and the right hand pushing the two eyetubes at the same time, and changing the position of any one of the eyetube at will.|
Eye Tube AngleClose Λ
|Usually the Microscope Eyetube is 45°, some is 30°, Tiltable Eyetube Angle design of a microscope is also known as the ergonomics microscope.|
0-30° or 0-45° is an ergonomic design. When the mechanical tube length / focal length of the tube of the microscope is relatively big, the microscope is relatively high, and the user's height or the seat of the work desk is not suitable, long-term use of microscope may cause sitting discomfort.
Eyepiece tube with variable angle can freely adjust the angle without lowering the head. Especially when it is close to 0 degree and the human eye is close to horizontal viewing, long-time or long-term use can avoid fatigue damage to the cervical vertebra.
Erect/Inverted ImageClose Λ
|After imaging through a set of objective lenses, the object observed and the image seen by the human eye is inverted. When the observed object is manipulated, move the specimen or object, the image will move in the opposite direction in the field of view. Most of the biological microscopes are reversed-phase designs.|
When needing to operate works with accurate direction, it is necessary to design it into a forward microscope. Generally stereo microscopes and metallurgical microscopes are all of erect image design.
When observing through the camera and display, the erect and inverted image can be changed by the orientation of the camera.
360° Degree RotatableClose Λ
|The eyepiece of the microscope can have different viewing or observing directions. When the position of the microscope is uncomfortable, the direction of the eyepiece tube of the microscope can be adjusted, to facilitate observation and operation.|
Placement method of different viewing angles of the microscope:
General direction: the support column is behind the object to be observed
Reverse direction: the support column is in front of the object to be observed
Lateral direction: the support column is on the side of the object to be observed
Rotating eyepiece tube, different microscopes may have different methods, for some, the direction is confirmed when installing the eyepiece tube of the microscope, for some, by rotating the body of the microscope, and for some, by rotating the support member on the support or holder of the microscope.
Interpupillary AdjustmentClose Λ
|The distance between the two pupils of the human eye is different. When the image of exit pupil of the two eyepieces of the microscope are not aligned with the entry pupil of the eye, the two eyes will see different images, which can cause discomfort.|
Adjust the distance between the two eyepieces, to accommodate or adapt to the pupil distance of the observer's eyes. The adjustment range is generally between 55-75mm.
Image Port Switch ModeClose Λ
|The third eyepiece splitting in the trinocular microscope is to borrow one of the two sets of eyepiece optical paths as the photographic light path. The beam split prism or beam splitter can reflect part of the image light to the eyepiece, and part passes through to the third eyepiece photographic light path, such a trinocular microscope is called trinocular simultaneous imaging microscope, or true-trinocular.|
The beam split prism or beam splitter of the trinocular simultaneous imaging microscope or true-trinocular often has different splitting modes, such as 20/80 and 50/50, etc. Usually, the former is the luminous flux ratio of the eyepiece optical path, and the latter is the luminous flux ratio of the photographic optical path.
The advantage of true-trinocular is that, the real three optical paths can be imaged at the same time, and are not affected by the simultaneous use of the eyepiece observation and the photographic optical path (display). The disadvantage is that, because of the reason of the splitting, the image light of the photography is only a part. In theory, the image effect will be affected, and the effect is more obvious in the binocular eyepiece observation. If viewed closely, one will find that the eyepiece of the light path is relatively dark. However, in the current optical design and materials, the impact on the actual work is not very big, especially in the observation of low magnification objective lens, it has basically no effect at all, and therefore used by many people.
Adjustable EyepieceClose Λ
|The adjustable eyepiece is between the lens of the eyepiece and the focal plane, with distance adjustable device.|
For most people, their two eyes, the left and the right, have different vision. For adjustable eyepieces, the eyepoint height of the eyepiece can be adjusted to compensate for the difference in vision between the two eyes, making the image in the two eyes clear and consistent.
The range of adjustment of the general eyepiece is that the diopter is plus or minus 5 degrees, and the maximum difference between the two eyepieces can reach 10 degrees. Before use, it is generally necessary to adjust both eyepieces to the initial position where the scale is displayed as 0, which is used as a baseline to facilitate up and down adjustment.
The reticle position of the eyepiece is generally 10mm below the fixed position of the eyepiece tube. Because the vision of each person is different, some people may not be able to see the reticle clearly. For adjustable eyepiece, the height of the reticle position can be adjusted to make the reticle and the observed object clear at the same time, this is the advantage of adjustable eyepiece that mounts the diopter adjustment on the eyepiece tube compared with non-adjustable eyepiece.
When non-adjustable eyepiece is equipped with a reticle, if the diopter is adjusted, the reticle will rotate accordingly, thereby affecting the position of the measurement. For adjustable eyepiece, when its diopter is adjusted, its reticle does not rotate.
Eyepiece Optical MagnificationClose Λ
|Eyepiece optical magnification is the visual magnification of the virtual image after initial imaging through the eyepiece. When the human eye observes through the eyepiece, the ratio of the tangent of the angle of view of the image and the tangent of the angle of view of the human eye when viewing or observing the object directly at the reference viewing distance is usually calculated according to 250 mm/focal length of eyepiece.|
The standard configuration of a general microscope is a 10X eyepiece.
Usually, the magnification of the eyepiece of compound microscope is 5X, 8X, 10X, 12.5X, 16X, 20X.
As stereo microscope has a low total magnification, its eyepiece magnification generally does not use 5X, but can achieve 25X, 30X and other much bigger magnification.
Eyepiece Field of ViewClose Λ
|The eyepiece field of view is the diameter of the field diaphragm of the eyepiece, or the diameter of the image plane of the field diaphragm imaged by the field diaphragm.|
The diameter of a large field of view can increase the viewing range, and see more detail in the field of view. However, if the field of view is too large, the spherical aberration and distortion around the eyepiece will increase, and the stray light around the field of view will affect the imaging effect.
Eyepoint TypeClose Λ
|Eye point refers to the axial distance between the upper end of the metal frame of the eyepiece and the exit of pupil.|
The exit of pupil distance of high eyepoint eyepiece is farther than that of the eye lens of the ordinary eyepiece. When this distance is greater than or equal to 18mm, it is a high eyepoint eyepiece. When observing, one does not need to be too close to the eyepiece lens, making it comfort to observe, and it can also be viewed with glasses. Generally, there is a glasses logo on the eyepiece, indicating that it is a high eyepoint eyepiece.
Objective Working DistanceClose Λ
|The objective working distance is the vertical distance from the foremost surface end of the objective of the microscope to the object surface to be observed.|
Generally, the greater the magnification, the higher the resolution of the objective, and the smaller the working distance, the smaller the field of view. Conversely, the smaller the magnification, the lower the resolution of the objective, and the greater the working distance, and greater the field of view.
High-magnification objectives (such as 80X and 100X objectives) have a very short working distance. Be very careful when focusing for observation. Generally, it is after the objective is in position, the axial limit protection is locked, then the objective is moved away from the direction of the observed object.
The relatively greater working distance leaves a relatively large space between the objective and the object to be observed. It is suitable for under microscope operation, and it is also easier to use more illumination methods. The defect is that it may reduce the numerical aperture of the objective, thereby reducing the resolution.
Objective Screw ThreadClose Λ
|For microscopes of different manufacturers and different models, the thread size of their objectives may also be different. |
In general, the objective threads are available in two standard sizes, allowing similar objectives between different manufacturers to be used interchangeably.
One is the British system: RMS type objective thread: 4/5in X 1/36in,
One is metric: M25 X 0.75mm thread.
Boom StandClose Λ
|Boom stand is also called universal stand.|
It is a relatively large pole type stand. The height and length of the stand are big, and it can be freely adjusted in height, length and various angles. Its large weight ensures stable support and occupation of large space, but it can make the microscope free to move in a wide range with convenience. Boom stand is suitable for observing large objects.
The direction of boom stand is flexible, and when in use, various kinds of positions and methods can be adopted, such as front, side, and tilt etc., to facilitate the layout of the workbench. On the side of the crossbar of the boom stand, a 5/8-inch connecting hole is generally left for connecting various focusing mechanisms and microscopes.
The base of the boom stand usually only plays a fixing and supporting role. Under the observation of the microscope, it is an empty workbench, which can be used to place various platforms, work operating surfaces, and tools, etc., and can be freely combined into different working positions. When the base is large, the object to be observed can also be placed.
In industrial places, most of the working positions are fixed. Sometimes, in one working position, a lot of tools, equipment and instruments need to be placed.. Because the microscope is relatively large in size and takes up also a relatively bigger space, and not convenient to move back and forth, therefore for purpose of use, the boom stand can be placed in an appropriate position, and does not need to occupy the most commonly used work tables. When in use, the microscope can be moved over, and pushed to the side when not in use. This is very suitable for use in electronics factories, installation and maintenance, medical and animal anatomy, archaeology and other industries.
Boom stand generally does not have a fixed focusing device, and you can choose a variety of flexible accessories.
Because the stand needs to ensure flexibility, therefore there are many locking buttons in all directions. In any time after adjustment, it must be ensured that each knob is in a locked state to avoid sliding, tilting and flipping of the microscope, thereby damaging the microscope and the items on the workbench.
|Usually the universal joint is called E-Arm, i.e., Easy-Arm, also known as Universal Arm. Many people in the industry call it Bonder Arm, which refers to the components that connect the microscope on the COG Bonding Machine.|
At the tail of the E-arm there is a standard 5/8 inch (0.625 inch, 15.875mm) connector. The connector can be moved freely in both horizontal and vertical directions, and can also be fixed at an angular position in the vertical direction to facilitate microscope observation from different angles.
E-arm can be connected to various kinds of microscope stands with 5/8-inch adapters, such as boom stand, flexible arm etc. It is also possible to connect various kinds of microscopes by adding or replacing different adapters. Note that, in general, these stands themselves are not directly configured with this E-arm, and separate purchase is necessary.
Dia. 76mm Scope HolderClose Λ
|The 76mm stand scope holder is the most popular microscope body adapter size, suitable for stereo microscopes produced by most manufacturers.|
Place the microscope body in a 76mm scope holder, tighten with screws to avoid shaking when the microscope is in use.
Because this stand scope holder is very common, some special-sized microscopes can also borrow and use this stand, but only need a specific adapter to connect the microscope body with a diameter of less than 76mm.
Coupler/C-mount AdapterClose Λ
|Coupler/C-mount adapter is an adapter commonly used for connection between the C-adapter camera (industrial camera) and a microscope.|
Adjustable CouplerClose Λ
|On the coupler/C-mount-adapter, there is an adjustable device to adjust the focal length.|
Coupler for Microscope TypeClose Λ
|Different coupler/C-mount-adapters are suitable for different microscopes. For some, some adapter accessories need to be replaced. See the applicable range of each coupler/C-mount-adapter for details.|
Coupler MagnificationClose Λ
|Coupler magnification refers to the line field magnification of the coupler/C-mount-adapter. With different magnifications of the adapter lens, images of different magnifications and fields of view can be obtained. The size of the image field of view is related to the sensor size and the coupler/C-mount-adapter magnification.|
Camera image field of view (mm) = sensor diagonal / coupler/C-mount-adapter magnification.
For example: 1/2 inch sensor size, 0.5X coupler/C-mount-adapter coupler, field of view FOV (mm) = 8mm / 0.5 = 16mm.
The field of view number of the microscope 10X eyepiece is usually designed to be 18, 20, 22, 23mm, less than 1 inch (25.4mm). Since most commonly used camera sensor sizes are 1/3 and 1/2 inches, this makes the image field of view on the display always smaller than the field of view of the eyepiece for observation, and the visual perception becomes inconsistent when simultaneously viewed on both the eyepiece and the display. If it is changed to a 0.5X coupler/C-mount-adapter, the microscope image magnification is reduced by 1/2 and the field of view is doubled, then the image captured by the camera will be close to the range observed in the eyepiece.
Some adapters are designed without a lens, and their optical magnification is considered 1X.
For Camera Sensor SizeClose Λ
|For the size of the lens field of view of the coupler/C-mount-adapter, in the design process, the size of the camera sensor imaging target should be considered. When the field of view of the lens is smaller than the target plane of the camera, “black border” and “dark corner” will appear.|
The general microscope coupler/C-mount adapters are generally designed for the 1/2" camera targets. When a camera of 2/3 or larger target is used, the “dark corner” phenomenon will appear in the field of view. Especially, at present, DSLR cameras generally use large target plane design (1 inch full field of view), when used for microscopic photographing, the general DSLR camera coupler/C-mount adapter will have “black border”.
Generally, the “dark corner” that appears on the field of view is often that the center of the microscope and the camera are not aligned. Adjust the position of the screw on the camera adapter, or turn the camera adapter to adjust or change the effect.
C/CS-Mount CouplerClose Λ
|At present, the coupler/C-mount adapter generally adopts the C/CS-Mount adapter to match with the industrial camera. For details, please refer to "Camera Lens Mount".|
|After unpacking, carefully inspect the various random accessories and parts in the package to avoid omissions. In order to save space and ensure safety of components, some components will be placed outside the inner packaging box, so be careful of their inspection.|
For special packaging, it is generally after opening the box, all packaging boxes, protective foam, plastic bags should be kept for a period of time. If there is a problem during the return period, you can return or exchange the original. After the return period (usually 10-30 days, according to the manufacturer’s Instruction of Terms of Service), these packaging boxes may be disposed of if there is no problem.
|Microscope Optical Data Sheet|
|P/N||Objective||Objective Working Distance||Eyepiece|
|SZ17011141 (10X Dia. 22mm)||SZ17013421 (15X Dia. 16mm)||SZ17013621 (20X Dia. 12.5mm)|
|Magnification||Field of View(mm)||Magnification||Field of View(mm)||Magnification||Field of View(mm)|
|1. Magnification=Objective Optical Magnification * Body Magnification * Eyepiece Optical Magnification|
|2. Field of View=Eyepiece Field of View /（Objective Optical Magnification*Body Magnification）|
|3. The Darker background items are Standard items, the white background items are optional items.|
|Video Microscope Optical Data Sheet|
|1. Magnification=Objective Optical Magnification * Body Magnification * Coupler Magnification|
|Camera Image Sensor Specifications|
|No.||Camera Image Sensor Size||Camera image Sensor Diagonal|
|Digital Magnification Data Sheet|
|Image Sensor Size||Image Sensor Diagonal size||Monitor|
|Screen Size (15in)|
|Digital Zoom Function|
|1. Digital Zoom Function= (Screen Size * 25.4) / Image Sensor Diagonal size|
|Microscope Optical and Digital Magnifications Data Sheet|
|Objective||Coupler||Camera||Monitor||Video Microscope Optical Magnifications||Digital Zoom Function||Total Magnification||Field of View (mm)|
|PN||Magnification||PN||Magnification||Image Sensor Size||Image Sensor Diagonal size||Screen Size|
|1. Video Microscope Optical Magnifications=Objective Optical Magnification * Body Magnification * Coupler Magnification|
|2. Digital Zoom Function= (Screen Size * 25.4) / Image Sensor Diagonal size|
|3. Total Magnification= Video Microscope Optical Magnifications * (Screen Size * 25.4) / Image Sensor Diagonal size|
|4. Field of View (mm)= Image Sensor Diagonal size / Video Microscope Optical Magnifications|
|Product Instructions/Operation Manual||1pc|
|Packaging Type||Carton Packaging|
|Packaging Material||Corrugated Carton|
|Packaging Dimensions(1)||26x25x24cm (10.236x9.842x9.448″)|
|Packaging Dimensions(2)||67x32x18cm (26.378x12.598x7.087″)|
|Packaging Dimensions(3)||29x16x18cm (11.417x6.299x7.087″)|
|Inner Packing Material||Plastic Bag|
|Ancillary Packaging Materials||Expanded Polystyrene|
|Gross Weight||22.8kg (50.42lbs)|
|Minimum Packaging Quantity||1pc|
|Transportation Carton||Carton Packaging|
|Transportation Carton Material||Cardboard Box|
|Transportation Carton Dimensions(1)||26x25x24cm (10.236x9.842x9.448″)|
|Transportation Carton Dimensions(2)||67x32x18cm (26.378x12.598x7.087″)|
|Transportation Carton Dimensions(3)||29x16x18cm (11.417x6.299x7.087″)|
|Total Gross Weight of Transportation(kilogram)||22.80|
|Total Gross Weight of Transportation(pound)||50.42|
|Quantity of One Transportation Carton||3pc|